Dog Cardiac Troponin-I SPARCL™ Assay | CTNI-SP-4
- One Week Delivery
- 96 Wells
Troponin complex regulates striated muscle contraction. It is comprised of three subunits: troponin I, troponin C, and troponin T. Troponin I exists in three isoforms; one in fast-twitch, one in slowtwitch, and one in cardiac muscle. Cardiac troponin-I (cTnI) is significantly different from the skeletal muscle isoforms, allowing the development of cTnI specific immunoassays. After cardiac injury, cTnI is released into the blood from damaged muscle cells. Measurement of serum cTnI allows assessment of the extent of cardiac injury. It is widely used as a cardiac biomarker in in preclinical and veterinary research.
PRINCIPLE OF THE ASSAY
The dog cTnI SPARCLâ¢1 (Spatial Proximity Analyte Reagent Capture Luminescence, ref 1) assay uses two different affinity purified cTnI-specific antibodies that have been used in our preclinical cTnI ELISA kits2 since 2006. One is conjugated to horseradish peroxidase (HRP) and the other is conjugated to acridan, a chemiluminescent substrate. When HRP and acridan conjugated cTnI antibodies bind to cTnI they are brought into close proximity. With the addition of hydrogen peroxide, HRP catalyzes oxidation of proximal acridan molecules causing a flash of chemiluminescence. Acridan conjugated antibodies distant from HRP produce no signal. This principle allows the development of a homogeneous assay that allows rapid determination of cTnI concentrations.
The HRP and acridan conjugated antibodies provided with the kit are mixed with standards and diluted samples in wells of the 96-well white SPARCLâ¢ plate provided with the kit3. After incubation for 30 minutes on a shaker at 25Â°C and 150 rpm, the plate is placed into a luminometer. Trigger solution containing hydrogen peroxide is injected into each well and luminescence is immediately measured. The concentration of cTnI is proportional to luminescence and is derived from a standard curve.
MATERIALS AND COMPONENTS
Materials provided with the kit:
â¢ Anti-cTnI HRP stock Store â¤ -70Â°C
â¢ Anti-cTnI acridan stock Store â¤ -70Â°C
â¢ cTnI stock Store â¤ -20Â°C
â¢ Sample diluent; YD25-1, 25 ml
â¢ Conjugate diluent; CSD10-1, 10 ml
â¢ Trigger solution; TS11-1, 11 ml
â¢ White SPARCLâ¢ plate (12 x 8-well)
â¢ Clear untreated 96-well plate
Materials required but not provided:
â¢ Precision pipettes and tips
â¢ Polypropylene microcentrifuge tubes
â¢ Vortex mixer
â¢ Plate incubator/shaker
â¢ Luminometer capable of simultaneous injection/measurement
â¢ Curve fitting software
Store the HRP conjugate, acridan conjugate and haptoglobin stock at -70Â°C (they may be stored at -20Â°Cfor one week). The remainder of the kit should be stored at 2-8Â°C. The SPARCLâ¢ plate should be kept in a sealed bag with desiccant and antioxidant. The kit will remain stable for at least six months from the date of purchase, provided that the components are stored as described above
1. Please take the time to completely read all instructions before starting your assay. Contact us if you need clarification.
2. All reagents used in the assay should be allowed to reach room temperature (25Â°C) before use.
3. It is important that standards and samples be added to the SPARCLâ¢ plate quickly. If testing large numbers of samples, rather than pipetting standards and samples directly into the white SPARCLâ¢ plate using a single channel pipettor, we recommend the following. First, pipette an excess volume of standards and samples into appropriate wells of the clear 96- well plate. Then use an 8- or 12-channel multipipettor to quickly and efficiently transfer 50 ml aliquots to the appropriate wells of the white SPARCLâ¢ plate. The wells of the clear plate hold a maximum volume of 300 ml.
4. Follow the sequence of events below when running the assay.